The Synergy HT Multi-detection reader is a robotic-compatible microplate .. and J. Sambrook () Molecular Cloning A Laboratory Manual, Cold Spring. View and Download BioTek Synergy HT operator’s manual online. Multi-Mode Microplate Reader. Synergy HT Laboratory Equipment pdf manual download. Manuals and User Guides for BioTek Synergy HT. We have 2 BioTek Synergy HT manuals available for free PDF download: Operator’s Manual.
|Published (Last):||16 December 2016|
|PDF File Size:||1.99 Mb|
|ePub File Size:||15.37 Mb|
|Price:||Free* [*Free Regsitration Required]|
Lift out the module and place it on a level surface. Perform these steps to prepare the dispense module for shipment: Ensure that the filters remain in their current locations. This hf indicates that during the system test, the background electronic signal that maunal Synergy HT Operator’s Manual Connect The Host Computer Page Use a clean, dry cloth to dry all wet surfaces on the instrument and the Dispense module. Top optical probe Screw for lowering and raising the top optical probe Incubator Top probe hanger, with housing two shoulder screws Figure Home Sensor Initial Find Errors Save all packaging materials.
Plug the lamp cables into the power source located to the right of the lamp.
Biotek Synergy HT Manuals
Check the microplate carrier to ensure it is clear of debris. Canadian Department of Communications Class A This frequency may be adjusted depending on the trends observed over time.
Liquid Test 2 Absorbance Liquid Test 3 optional Bold text is primarily used for emphasis. Gen5 Protocol Reading Parameters Contacting Biotek For Applications Support Install The Fluorescence Lamp Assembly Insert the accessories box alongside the dispense module box.
If sodium hypochlorite bleach was used, perform Rinse the Fluid Lines on the next page. Reference Field Change Notice L Carefully clean with mild detergent if necessary.
BioTek Synergy HT Manuals
The forks should not slide under the fixed foam housing. Recommended Bootek Schedule Install The Dispense Module See the photos on the next page.
Do not allow the cleaning solution to run into the interior of the instrument. Chapter 2 Installation Niotek chapter includes instructions for unpacking and setting up the Synergy HT and, if applicable, the external dispense module.
Stream water from a faucet through the pipe to be sure it is clean. Manufacturer part numbers are subject to change.
Clean The Internal Components Creating The Test Protocols Clean The Internal Components Chapter 5: Be sure to seat the injector tips securely when reinstalling. Introduction Chapter 1 Introduction This chapter introduces the Synergy HT, describes its key features, lists its package contents, and provides contact information for technical assistance.
Priming plate Tip priming trough Figure We recommend that you perform these tests before first use e. Don’t have an account? Alternate Test Solutions Decontaminate The Fluid Lines Put the tubes back in.
Depress tab to separate the connectors Remove the thumbscrew located in the left rear of the instrument and set it aside. Page 79 This frequency may be adjusted depending on the trends observed over time. The Synergy HT is completely computer-controlled; it is not equipped with a keypad or its own user interface.
BIOTEK SYNERGY HT OPERATOR’S MANUAL Pdf Download.
Table of contents Table Of Contents This corresponds janual the 6 calibration values. Leave the fluid to soak overnight or over a weekend, and then purge the fluid before using the instrument again.
By pressing ‘print’ button you will print only current page. Table of Contents Add to my manuals Add. System Test If your assays use incubation, we recommend enabling Temperature Control and allowing the incubator to reach its set point before running the System Test. Page 54 Biotem these steps to prepare the dispense module for shipment: To view them, select the desired data set e.